|Home||Science||Detector||Collaboration||Publications & Talks||Events||LSM||internal|
High resistivity, fully depleted CCDs
The detectors were fabricated from n-type, high-resistivity (>10,000 Ω cm) silicon wafers, and are fully depleted (i.e., active over their full volume) by applying a potential (≥ 40 V) to a thin back-side contact (Fig. 1.b).
The 15 μm x 15 μm pixels have a conventional 3-phase polysilicon gate structure to hold and transfer the charge, arranged in an array of 4k x 4k pixels.
In the serial readout process, the charge is transferred vertically from pixel to pixel along each column by appropriate clocking of the 3-phase gates until it reaches the last row (“serial register”); higher frequency clocks move the charge horizontally until the end of the serial register where the on-chip charge-to-voltage amplifier is located. The corresponding voltage signal is then processed to determine the pixel charge. A readout noise of ≈ 1.6 electrons has been achieved at SNOLAB.
CCDs as Dark Matter detectors
The ionized charge diffuses transversely as it is drifted toward the pixel gates, with a spatial variance that is proportional to the transit time (Fig. 2.b). Hence, there is a positive correlation between the lateral diffusion of the collected charge on the pixel array and the depth of the interaction, which can be used for three-dimensional position reconstruction.
Background identification and rejection
A truly unique capability of DAMIC is that background can be identified and rejected as spatially correlated events occurring at different times. An example is shown in Fig. 3.b, with three clusters (two electrons and one alpha) detected in the same location but separated in time by several days. The probability for this to occur by chance is negligible. Rather, the observation is consistent with a 210Pb nucleus undergoing the decay sequence 210Pb (T1/2= 22.3 y, β) → 210Bi (T1/2=5.0 days, β) → 210Po (T1/2= 138 days, α). A product of the radon decay chain, 210Pb is a major source of surface background in DM experiments caused by material exposure to air.
This background rejection capability is also particularly relevant for 32Si, a naturally occurring isotope in silicon that is not removed by chemical purification, which is expected to be the dominant background contribution in the current generation of silicon detectors. DAMIC identifies and rejects by spatial correlation the decay sequence 32Si (T1/2= 150 y, β) → 32P (T1/2= 14 days, β).
Skipper CCDs with single-electron resolution
The crucial innovation in DAMIC-M is the integration on these large devices of a novel on-chip readout scheme, where sub-electron noise is achieved by a non-destructive, multiple measurement of the pixel charge. In the so-called “skipper” readout, the charge of a pixel is measured N times, and by taking the average of these measurements the white noise is reduced by √N. Most importantly, the effect of low-frequency 1/f noise is drastically reduced, since the integration time of each measurement is much shorter than in the conventional readout.
This readout has been successfully implemented in a 6 Mpixel DAMIC-M prototype CCD. Its pixel charge distribution is shown in Fig. 4, with well-separated peaks corresponding to zero, one, two, etc. electrons.
With a noise of only ≈ 0.1 e-, DAMIC-M will detect with high-resolution a single electron. Thus, it will be sensitive to extremely small energy transfers from a DM-electron interaction, since 1.1 eV (band gap energy) is the threshold to produce an electron-hole pair in silicon (with 3.7 eV required on average).
Even with single-charge resolution, a signal may remain hidden in the fluctuations of the detector’s dark current. DAMIC CCDs have the lowest dark current ever measured in a silicon detector, ≤ 2x10-22 A/cm2 at an operating temperature of 140 K.
DAMIC-M at the Laboratoire Souterrain de Modane
A conceptual design of DAMIC-M at is shown in Fig. 5. A cryostat vessel houses the CCDs, closely packed in a box surrounded by infrared (IR) shields. Cooling is provided by a copper cold finger connected to a liquid nitrogen dewar located outside the shielding. The vacuum cryostat, IR shields, CCD box and cold finger are made of electroformed copper to minimize radiogenic backgrounds. The cryostat is enclosed in a 20-cm-thick lead shielding to stop high-energy radiogenic photons from the external environment. Standard lead contains quantities of 210Pb (T1/2 = 22 years, β) whose decay chain products may generate penetrating bremsstrahlung photons escaping the shield. For this reason, the innermost 5 cm of the shield are made of ancient lead available at the LSM, recovered from a Roman ship sunk in Brittany in 400 A.C. The entire lead shield is enclosed by 40-cm-thick polyethylene to efficiently stop external neutrons.